Open as PDF
CHAPTER 11: PROGRAMMING GROUP 2 - METER CALIBRATION GROUP 2, FUNCTIONS 0–2
EPM 5300 SERIES ADVANCED POWER METERS – INSTRUCTION MANUAL 11–87
a. Enter Group Level of Programming Mode
(see Chp. 8).
b. Press MAX/MIN/LIMITS until 2. appears in upper
c. Press AMPS to activate the Group.
A one digit password is required to continue.
d. Press VOLTS until 5 appears.
e. Press AMPS to select.
a. Press MAX/MIN/LIMITS until 22. appears in upper
b. Press AMPS to activate. (Calibration begins with Amps
A, then moves to B and C).
Middle and lower displays blank.
New value moves to middle display.
Lower display indicates the calibrated reading after
To enter the half scale value in lower display,
a. Press VOLTS to select digits.
b. Press AMPS to store and proceed to B or C within the
c. Press MAX/MIN/LIMITS to end calibration procedure.
See Chapter 14 to Exit.